Guidelines for Tooth Slooth Fracture Detector (white) and Tooth Slooth II-Crown Seater (blue)
Rinse Tooth Slooth under running water or use a mild detergent to remove any debris.
Place Tooth Slooth inside a sterilization pouch.
CYCLE TEMPERATURE TIME DRY TIME
Prevac 270ºF/132ºC 4 minutes *1-30 minutes
Gravity 270ºF/132ºC 10 minutes *1-30 minutes
*Dry time not to exceed 30 minutes
Note: Dry heat setting is NOT recommended as it will cause the instruments to melt.
To dry, use the recommended drying time the sterilizer.
Instruments to remain bagged until ready for use.
Use cleaning and sterilization process that meet ADA and CDC guidelines. Due to the variety of makes and models of sterilizers, the recommended settings may not be applicable. It may be necessary to refer to your specific brand of sterilizer for their recommended time and temperature setting for plastics (polypropylene).